MIC TECH’S NEW MIC-50 biological microscope is designed to present a safe, comfortable and efficiency observation experience for users. With perfectly performed structure, high-definition optical image and simple operating system, MIC-50 realizes professional analysis, and meets all the needs of research in scientific, medical and other fields.
Large stage with adjustment in either hand In order to correct the shortcoming of horizon guide rail, new stage is designed with double-way linear driving mechanism. This change protects the stage from overload at the end of both rails, improves the rigidity and performance of the stage.
The handle of the stage can be set at each side based on users’ preference. The X, Y biaxial adjustment are designed with low position for comfortable operation.
Two slices can be hold on the stage by using damping-type double clips, easy for comparative study. Moving range: 80mm X55mm; precision: 0.1mm Processed with special craft, the surface of the stage is anti-corrosive and anti-friction. The platform with an arc transition design reduces the stress concentration and damage from impact.
Light barrier, which can effectively reduce the stray light, improves the image contrast in fluorescence observation.
Modular frame, improve the system compatibility MIC-50 with modularization design, separated cross arm and main body, improves the system compatibility of biological and fluorescence frame.
Highly sensitive coaxial coarse and fine adjustment Coaxial adjustment adopts double-stage driving, with adjustable tension tightness and upper limit stop, coarse range is 25mm and fine precision is 1μm. Not only accurately focus but also precision measurement is available.
Viewing head with multi-splitting ratio MIC-50 viewing head is designed of multiple options for splitting ratio. with wide beam imaging system, 26.5mm super wide filed of view is available.
Trinocular head with inverted image, splitting ratio Binocular: Trinocular=100:0 or 20:80 or 0:100. Except for concentrating 100% light to eyepiece tube or camera tube, there is another option with 20% light to eyepiece tube and 80% to camera tube, so that eyepiece observation and image output can be available at the same time.
Trinocular head with erect image, splitting ratio Binocular:Trinocular= 100:0 or 0:100. The moving direction of samples is as the same as observed.